Helping to protect your most vulnerable patients from leak-related complications
The new Puritan Bennett™ 980 ventilator was designed with the challenges of safe and effective ventilation in mind. Automatically detecting and compensating for fluctuating leak sizes, Puritan Bennett™ Leak Sync software helps clinicians manage patients’ work of breathing.
Breathing circuit leaks can cause a ventilator to erroneously detect patient inspiratory efforts (called autotriggering) or delay exhalation in pressure support. Patient interfaces, such as masks and uncuffed endotracheal tubes, are particularly prone to significant leaks. Inaccurately declaring inspiration or exhalation can result in patient-ventilator asynchrony and increased work of breathing.
Leak Sync software accurately quantifies instantaneous leak rates in adults, pediatric and neonatal patients during invasive or noninvasive ventilation. It compensates for leaks up to 65 L/min in adults, 40 L/min in pediatric patients and 15 L/min in neonates.
Puritan Bennett™ Leak Sync Software Highlights
Download the Leak Sync Software Guide
Read the Leak Sync software quick reference guide for the Puritan Bennett 980 ventilator system.
The Puritan Bennett™ Leak Sync software is similar to the Leak Compensation software on the Puritan Bennett™ 840 ventilator, but has been expanded to be used with VC+ and VS. Leak compensation software has been proven to synchronize faster than other leading ventilators to increasing and decreasing leaks.1
The Puritan Bennett™ 980 ventilator with Leak Sync technology is able to:
- Automatically compensate for leaks up to 65 L/min in adults, 40 L/min in pediatric patients and 15 L/min in neonates, helping to reduce trigger and cycle asynchronies
- Quantify and compensate for new leaks or changes in leak rate within approximately three breaths2
- Display the percentage of volume lost to the leak (% LEAK), leak rate (LEAK L/min) and leak volume directly on the graphical user interface (GUI)
Leak Sync software works with the following breathing strategies:
|Volume Control||Volume Control|
|Pressure Control||Pressure Control|
|Pressure Support||Pressure Support|
1. Leak Sync software is an enhancement to the Leak Compensation software found in: Oto J, Chenelle CT, Marchese AD, Kacmarek RM. A comparison of leak compensation during pediatric non-invasive positive pressure ventilation; a lung model study.Respir Care. 2013;58(12):2027-2037.
2. Puritan Bennett™ 980 Ventilator Operations Manual
|Predicted body weight (PBW)||0.3 kg (0.66 lb) to 7 kg (15 lb)|
|Modes||Assist Control (A/C), Synchronized Intermittent Mandatory Ventilation (SIMV), Spontaneous (SPONT), BiLevel, Continuous Positive Airway Pressure (CPAP)|
|Mandatory breath types||Volume Control (VC), Pressure Control (PC) and Volume Control Plus (VC+)|
|Spontaneous breath types||Pressure Support (PS), Volume Support (VS)|
|Ventilation type||Invasive and Noninvasive (NIV)|
|Pressure support (PSUPP)||0 cmH2O to 70 cmH2O|
|Flow Pattern||Square or descending ramp|
|Rise time %||1% to 100%|
|Expiratory sensitivity (ESENS)||1% to 80%|
|Tidal volume (VT)||2 mL to 315 mL|
|Respiratory rate (ƒ)||1.0 1/min to 150 1/min|
|Peak inspiratory flow (V̇MAX)||1 L/min to 30 L/min|
|Plateau time (TPL)||0.0 to 2.0 seconds|
|Inspiratory pressure (PI)||5 to 90 cmH2O|
|Inspiratory Time (TI)||0.2 to 8.0 seconds|
|I:E ratio||1:299 to 4:1|
|Expiratory time (TE)||≥ 0.20 seconds|
|Trigger types||Flow-triggering (V̇-TRIG)|
|Pressure sensitivity (PSENS)||0.1 cmH2O to 20 cmH2O|
|Flow sensitivity (V̇SENS)||0.1 L/min to 10 L/min|
|O2%||21% to 100%|
|Positive end expiratory pressure (PEEP)||0 cmH2O to 45 cmH2O|
|Apnea ventilation mandatory type||PC, VC|
|Apnea flow pattern||Square or descending ramp|
|Apnea peak inspiratory flow (V̇MAX)||1.0 L/min to 30 L/min|
|Apnea tidal volume (VT)||≥3 mL|
|Apnea inspiratory pressure (PI)||5 cmH2O to 90-PEEP cmH2O|
|Apnea interval (TA)||10 to 60 seconds or OFF in CPAP|
|Apnea respiratory rate (ƒA)||2.0 1/min to 40 1/min|
|Neonatal default||20 1/min|
|Apnea O2%||21% to 100% O2|
|Apnea I:E ratio||≤ 1.00:1|
|Apnea inspiratory time (TI)||0.20 to 8 seconds|
|Apnea expiratory time (TE)||0.20 to 59.8 seconds|
|Disconnect sensitivity (DSENS)||0 L/min to 15 L/min or 20% to 95% or OFF (if Leak Sync software not enabled)|
|Humidification type||Heat-moisture exchanged (HME), non-heated expiratory tube, heated expiratory tube|
|Humidifier volume||100 mL to 1,000 mL|
Airleaks can make safe and effective neonatal ventilation a challenge.
Clinicians strive to support healthy gas exchange and maintain clinical stability in NICU babies, while also protecting them from lung injury.1,2 Gas leaks around the endotracheal or tracheostomy tube can make this a challenge. Considerable leaks around the airway interface are common in newborns with the use of uncuffed endotracheal and tracheostomy tubes and airway masks.3
When breathing circuit leaks are not managed effectively, they can lead to4,5
- Inappropriate breath triggering and cycling
- Inadequate ventilation
- Inaccurate patient data
Detecting and responding to leaks is an important initiative that can improve treatment and reduce the infant’s work of breathing.6
The Puritan Bennett™ 980 neonatal ventilator was designed to help protect your most vulnerable patients from leak-related complications. Puritan Bennett™ Leak Sync software is available to automatically detect and compensate for fluctuating leak sizes, resulting in more accurate delivered tidal volumes. The software adjusts the effective trigger sensitivity in the presence of leaks, helping to reduce the incidence of auto-triggering and manage the baby’s inspiratory work of breathing.
By effectively managing airleaks during both invasive and noninvasive ventilation, the Puritan Bennett 980 neonatal ventilator can help clinicians provide the right ventilation, supporting the effort to keep babies safe and comfortable.7
- Mahmoud RA, Proquitté H, Fawzy N, Bührer C, Schmalisch G. Tracheal tube airleak in clinical practice and impact on tidal volume measurement in ventilated neonates. Pediatr Crit Care Med. 2011;12(2):197-202.
- Habre W. Neonatal ventilation. Best Pract Res Clin Anaesthesiol. 2010;24(3):353-364.
- Finholt DA, Henry DB, Raphaely RC. Factors affecting leak around tracheal tubes in children. Can Anaesth Soc J. 1985;32(4):326-329.
- Main E, Castle R, Stocks J, James I, Hatch D. The influence of endotracheal tube leak on the assessment of respiratory function in ventilated children. Intensive Care Med. 2001;27(11):1788-1797.
- Bougatef A. Neonatal mechanical ventilation. In: Gullo A. Anaesthesia Pain Intensive Care and Emergency Medicine. Milan, Italy: Springer. 2005:73-81.
- Oto J, Chenelle CT, Marchese AD, Kacmarek RM. A comparison of leak compensation during pediatric non-invasive positive pressure ventilation; a lung model study. Respir Care. 2013;58(12):2027-2037.
- Grossbach I, Chlan L, Trach MF. Ventilatory support and management of patient- and ventilator-related responses. Crit Care Nurse. 2011; 31(3):30-44.
Puritan Bennett™ 980 Ventilator Product Tour
Take a tour of the new ventilator to learn more about features, specifications and resources.
Start the Tour
Puritan Bennett™ Leak Sync Software Quick Reference Guide
See the Leak Sync Software Quick Reference Guide for the Puritan Bennett™ 980 Ventilator System.
The Puritan Bennett™ 980 Ventilator: Clinical Applications Lesson Plan
View the Leak Sync software overview presentation.
Puritan Bennett™ Leak Sync Software GUI Demonstration Video
See a brief video of Leak Sync software in use.