Professionisti del settore sanitario
QUESTO SITO UTILIZZA COOKIE
Con un browser aggiornato, l'esperienza sul sito Web di Medtronic sarà migliore. Aggiorna il mio browser adesso.
Si, sono un operatore sanitario
I contenuti presenti nella pagina che segue contengono informazioni rivolte esclusivamente agli OPERATORI SANITARI, in quanto si riferiscono a prodotti rientranti nella categoria dei dispositivi medici che richiedono l’impiego o l’intervento da parte di professionisti del settore medico-sanitario.
Professionisti del settore sanitario
Medtronic SureScan® pacing systems are MR Conditional and as such are designed to allow patients to undergo MRI under the specified conditions for use. A complete SureScan pacing system, which consists of an approved combination (see www.mrisurescan.com) MRI SureScan device with SureScan lead(s), is required for use in the MRI environment. Consult the device manuals to ensure all system components are MR Conditional.
Medtronic SureScan pacing systems are indicated for the following:
Note: the Advisa DR MRI pacing system includes the following additional indication:
The dual chamber systems are also indicated for dual chamber and atrial tracking modes in patients who may benefit from maintenance of AV synchrony. Dual chamber modes are specifically indicated for treatment of conduction disorders that require restoration of both rate and AV synchrony, which include:
Antitachycardia pacing (ATP) is indicated for termination of atrial tachyarrhythmias in bradycardia patients with one or more of the above pacing indications.
Medtronic SureScan pacing systems are contraindicated for:
There are no known contraindications for the use of pacing as a therapeutic modality to control heart rate. The patient’s age and medical condition, however, may dictate the particular pacing system, mode of operation, and implant procedure used by the physician.
Changes in patient’s disease and/or medications may alter the efficacy of the device’s programmed parameters. Patients should avoid sources of magnetic and electromagnetic radiation to avoid possible underdetection, inappropriate sensing and/or therapy delivery, tissue damage, induction of an arrhythmia, device electrical reset, or device damage. Do not place transthoracic defibrillation paddles directly over the device. Use of the device should not change the application of established anticoagulation protocols.
Patients and their implanted systems must be screened to meet the MRI Conditions of Use. Do not scan patients who do not have a complete Medtronic SureScan pacing system consisting of a SureScan device and two SureScan leads; patients who have broken, abandoned or intermittent leads; or patients who have a lead impedance value of < 200 Ω or > 1,500 Ω. Do not scan patients with a SureScan pacing system implanted in sites other than the left or right pectoral region.
Potential complications include, but are not limited to, rejection phenomena, erosion through the skin, muscle or nerve stimulation, oversensing, failure to detect and/or terminate arrhythmia episodes, acceleration of tachycardia, and surgical complications such as hematoma, infection, inflammation, and thrombosis. Potential lead complications include, but are not limited to, valve damage, fibrillation, thrombosis, thrombotic and air embolism, cardiac perforation, heart wall rupture, cardiac tamponade, pericardial rub, infection, myocardial irritability, and pneumothorax. Other potential complications related to the lead may include lead dislodgement, lead conductor fracture, insulation failure, threshold elevation, or exit block. The SureScan system has been designed to minimize potential complications in the MRI environment. Potential MRI complications include, but are not limited to, lead electrode heating and tissue damage resulting in loss of sensing or capture or both, or induced currents on leads resulting in continuous capture, VT/VF, and/or hemodynamic collapse.
See the device manuals before performing an MRI Scan for detailed information regarding the implant procedure, indications, MRI conditions of use, contraindications, warnings, precautions, and potential complications/adverse events. For further information, call Medtronic at 1 (800) 328-2518 and/or consult Medtronic’s website at www.medtronic.com.
Caution: Federal law (USA) restricts these devices to sale by or on the order of a physician.
Medtronic Italia S.p.A. Società soggetta ad attività di direzione e coordinamento da parte di Medtronic Inc.
Cap. soc. € 1.200.000,00 - Registro Imprese N. 281327 Tribunale Milano - R.E.A. N. 1275682 - Cod. Fisc./P.IVA 09238800156
Sede Legale e Uffici Via Varesina, 162 Edificio Raimondi - 20156 Milano