Your browser is out of date
With an updated browser, you will have a better Medtronic website experience. Update my browser now.
For product security updates, notification and documentation, please see the following:
Security vulnerabilities were identified related to the proprietary Medtronic Conexus™ radio frequency wireless telemetry protocol in some Medtronic products.
Security vulnerabilities were identified in Medtronic’s MiniMed™ Paradigm™ family of insulin pumps and corresponding remote controller. (03-13-19 | 08-07-18)
Security vulnerabilities were identified in Medtronic’s 9790 CareLink Programmer, 2090 CareLink Programmer and 29901 CareLink Encore Programmer
Security vulnerabilities were identified in the CareLink™ 2090 Programmer and accompanying software distribution network. (10-11-2018 | 02-27-2018)
Security vulnerabilities were identified in Medtronic’s MyCareLink™ Patient Monitor (model
numbers 24950 and 24952).
Security vulnerabilities were identified in the MyCareLink Patient Monitor that could allow privileged access to the operating system and visibility to product development code.
Security vulnerabilities were identified related to the N'Vision 8840 physician programmer and its removable compact flash application card.
Medtronic impact and response to the widely-publicized cybersecurity vulnerabilities known as Spectre and Meltdown, which affected many computer and mobile device processors in early 2018.
Medtronic impact and response to the WannaCry cyber attack affecting a number of organizations worldwide in spring 2017.