The Capnostream™ 20p patient monitor is built on a legacy of proven performance. For nearly two decades, clinicians have relied on Microstream™ technology for an accurate, continuous view of ventilation adequacy on intubated and non-intubated patients, from neonate to adult. The new Capnostream™ 20p monitor, with the Apnea-Sat Alert feature, alerts caregivers to repetitive patterns of apneas and oxygen desaturation, revealing a more complete picture of a patient’s respiratory status.
|Order Code||Description||Unit of Measure||Quantity|
|CS08651-02||Capnostream™ 20p Bedside Monitor with Apnea-Sat Alert Algorithm||Each||1|
|Power and Connectivity Accessories, and Spare Parts|
|010516||Paper (6 rolls). Paper fits Capnostream's integrated printer.||Box||1|
|010713||Mounting Adaptor Plate (Vesa). Used for mounting Capnostream to GCX roll stands and other mounting assemblies.||Each||1|
|011782||Quick Release Pole Clamp. Clamp fits pole mount (pole diameter 0.75 in [19 mm] to 1.5 in [38 mm]) or rail mount (rail size 10 mm [0.39 in] x 25 mm [0.98 in]).||Each||1|
|011892||Mounting Adapter for Bernoulli Transmitter. Used with a Bernoulli/Oxinet hospital system.||Each||1|
|011149||Nurse Call Cable. Cable length 3.5 meters.||Each||1|
|014256||US Power Cord (3 meters)||Each||1|
|014255||European Power Cord (3 meters)||Each||1|
|CS08807||Power Cable Strain Relief||Each||1|
|010543||Fuse, 3.14 amp/250 volt||Package||2|
Wireless Bridge Adapter (Bernoulli-compatible).
Supports all Nellcor™ OxiNet™ technology bridges:
– Moxa TCC-80 (DB 9) RS-232 to 422 Adapter
Quick Release Pole Clamp Kit.
Clamp ts pole mount (0.75”-1.5”/19 mm - 38 mm) or rail mount (0.39”/10 mm x 0.98”/25 mm).
Note: Never use a temporary mount when a falling Capnostream™ 20p monitor could cause patient injury.
Vesa Mounting Adapter Plate.
For mounting Capnostream™ 20p monitor on any standard Vesa 75 x 75 mounting system as required by various standard industry mounting solutions.
Roll Stand, Handle and Basket.
Features tilt/swivel adjustments. Post height is 38”/96.5 cm, overall height with casters, oor to mounting plate is 43”/109.2 cm. Mounting plate sold separately (010713).
Wall Channel, Vertical.
Standard channel. Channel permits monitor to be set anywhere in the channel.
Wall Channel, Vertical with Pivot Arm.
12” pivot arm length. Pivots at the slide for lateral adjustability.
|Unit dimensions||167 mm. (h) x 220 mm. (w) x 192 mm. (d)|
|(6.6 in. (h) x 8.7 in. (w) x 7.6 in. (d))|
|Unit weight||3.6 kg.|
|Packaged dimensions||315 mm. (h) x 340 mm. (w) x 285 mm. (d)|
|12.4 in. (h) x 13.4 in. (w) x 11.2 in. (d)|
|Input Voltage 100-240 V, 50/60 Hz, ± 10%.|
|Units: mmHg, kPa or Vol%|
|Measurement range etCO2, FiCO2 0-150 mmHg|
|Respiratory Rate 0 bpm-150 bpm|
|Resolution 1 mmHg|
|Accuracy (±SD) etCO2, FiCO2 0-38 mmHg ± 2 mmHg. 39-150 mmHg ± (5% of reading + 0.08% for every 1 mmHg above 38 mmHg)|
|Respiratory rate: 0-70 bpm ±1 bpm, 71-120 bpm ±2 bpm, 121-150 bpm ±3 bpm|
|Flow rate 50 ml/minute, flow measured by volume|
|Waveform sampling 50 millisecond intervals|
|Response time 2.95 second (typical)|
|Initialization time 40 second (typical)|
|Calibration interval Initial calibration after 1,200 operating hours, then annually or after 4,000 operating hours, whichever comes first|
|Nellcor™ Pulse Oximetry|
|Measurement range 0-100%|
|o SpO2 range 70%-100% ± 2 digits|
|o SpO2 range 0-69% unspecified|
|o SpO2 range 70%-100% ± 3 digits|
|o SpO2 range 0-69% unspecified|
|Pulse rate||o Range: 20-250 BPM|
|o Accuracy: ± 3 BPM|
|o Sat Sec Range: 10, 25, 50, 100|
Melissa Langhan, MD, reviews the importance of capnography to monitor pediatric patients in the emergency medicine environment.
†SARA evaluates the stability of the detected breath-to-breath intervals and adjusts (adapts) the measurement period accordingly, thus averaging out noise and transient fluctuations when required, providing a more realistic respiratory rate.
††SpO2 data from host required.
†††Apnea-Sat Alert is FDA cleared for adults 22 years of age or older.