The Nellcor™ N-85 portable pulse oximetry monitor with Microstream™ technology provides both accurate SpO2 and etCO2 values in a single, easy-to-use, handheld device.
Microcap™ handheld capnographs deliver an easy-to-use assessment of respiratory status, whether intubated or non-intubated, in any clinical setting.
See real results from hospitals using Microstream™ capnography monitoring.
Learn more about the outcomes associated with capnography monitoring during procedural sedation.
Find out how the design of sampling lines can impact accurate etCO2 measurement and quality waveforms.
Handheld monitors that are convenient for spot checks and continuous monitoring in a variety of settings, including EMS/ED, transport.
|Order Code||Description||Unit of Measure||Quantity|
|N85||Nellcor™ N-85 Hand-Held Capnograph/Pulse Oximeter with OxiMax™ Technology||Each||1|
|CS04178||Microcap™ Handheld Capnograph - North America/U.S.||Each||1|
|CS04179||Microcap™ Handheld Capnograph - Outside North America/Non-U.S.
|CS07779||Monitor Protective Boot||Each||1|
|CS03736||Rechargeable Battery Pack (replacement for original battery pack)||Each||1|
|CS04934||Medical Grade AC Adapter (replacement for original AC adapter) - US||Each||1|
|CS08610||Medical Grade AC Adapter (replacement for original AC adapter) - EU||Each||1|
|CS08505||12 Volt cable||Each||1|
|CS07143||D to A Converter (typically used to interface to sleep labs)||Each||1|
|CS04284||Communication Adapter Kit||Each||1|
|CS08644||MSM Interface Kit (Bernoulli, Oxinet)||Each||1|
|CS08642||Nurse Call Interface Kit||Each||1|
|Nellcor™ N-85 Hand-Held Capnograph/Pulse Oximeter with OxiMax™ Technology|
|Display Range||0-99 mmHg, 0-13.2 kPa and 0-13.0 Vol% at sea level|
|Sampling flow rate||50 ± 7.5 mL/min|
|Warm-up time||30 seconds typical, reaches steady-state|
|Accuracy||0 to 38 mm Hg ±2 mm Hg|
|39 to 99 mm Hg ±5% of reading|
|Interface||RS-232 interface at 9600 baud; full duplex serial interface via RJ-45 connector|
|Modes||Adult and Neonate|
|Alarms||Adjustable alarm limits for all parameters; audio level as defined in ISO 9703-2 standard|
|Display||Graphic LCD display (128 x 64 dots) with LED backlight (contrast and backlight control); two numeric fields, three digits each, using 7-segment LED; two-color LED alarm bar|
|Microcap™ Handheld Capnograph|
|Sampling flow rate||50 ml/ min|
|Warm-up time||30 seconds typical|
|Accuracy||0-38 mmHg ± 2 mmHg|
|39-99 mmHg ± 2 mmHg ± (5% of expected reading in mmHg +[0.08 x (expected reading in mmHg –39mmHg)])|
|Expected reading = (barometric pressure in mmHg) x (%CO2 (in Vol%)) x 0.97|
|Equivalent values for kPa and Vol%|
|Respiratory rate 0-150 breaths/min|
|Modes||Adult and Neonate|
|Alarms||Adjustable alarm limits for etCO2, FiCO2, and respiratory rate|
|Display||Graphic LCD (128 x 64 dots) with LED backlight, (contrast and backlight control)|
|Bright, seven-segment LED for numeric display of etCO2 and respiratory rate|
|Two color alarm bar|
|Interface||Nurse Call RS232 interface between monitor and remote systems|
|Bernoulli MSM Nurse Call System easily wired via stereo phone jack|
|Philips®* Patient Monitor Interface: Microcap® only - one parameter (v3.15 and higher)|