You just clicked a link to go to another website. If you continue, you will leave this site and go to a site run by someone else.
It is possible that some of the products on the other site not be licensed for sale in Canada.
Your browser is out of date
With an updated browser, you will have a better Medtronic website experience. Update my browser now.
By choosing to accept, you acknowledge that you are a Certified Healthcare Professional.
The patient scanning process refers to the steps that must be followed to help ensure patient safety during an MRI scan. On this page, you will find instructions for determining if a patient with a Medtronic SureScan™ System meets the criteria to receive an MRI, as well as instructions to follow before, during, and after an MRI scan.
|1.5T — MRI radiofrequency (RF) power||3T — MRI radiofrequency (RF) power|
Normal Operating Mode
First Level Controlled Operating Mode or Normal Operating Mode
*This includes visual and verbal contact with the patient and monitoring heart rate using instrumentation such as pulse oximetry or electrocardiography.
Download the SureScan Systems Patient Scanning Process PDF for a complete outline of the responsibilities of radiology and cardiology professionals.