The Microstream™ MicroPod™ external OEM module offers flexibility for OEM partners seeking to integrate market-leading Microstream™ capnography technology into a wide variety of existing medical devices.
The MicroPod™ external module configuration will help hospitals better manage their capital costs with a plug-and-play solution that can be shared between multiple devices that are MicroPod™-enabled.
MicroPod™ Microstream™ etCO2 technology is appropriate for medical devices, used on adult, pediatric and neonatal patients in a variety of care settings inside and outside the hospital — including patient monitors, defibrillators and ventilators.
Get detailed product specifications for the Microstream™ MicroPod™ OEM module.
Microstream™ technology offers a unique and highly specific measurement technology using Molecular Correlation Spectroscopy™.
Learn how quality sampling lines improve capnography accuracy.
|CO2 waveform, etCO2, FiCO2, Respiration Rate|
|Integrated Pulmonary Index™, Apnea-Sat Alert™ (Apneas Per Hour)|
|CO2 Partial Pressure (at sea level)||Accuracy|
|0-38 mmHg||± 2 mmHg|
|39-150 mmHg||± (5% of reading + 0.08x (reading – 39mmHG)|
|CO2 sampling flow rate||50 ml/min (+ 15 ml/min, -7.5 ml/min) flow measured by volume|
|Initialization time||Typically 30 seconds to obtain both reading and waveform.|
|At full accuracy when value first appears.|
|Respiration rate||0-150 breaths/min|
|Mode||Adult, pediatric, neonatal|
|Power requirements||1.2W (typical in normal operation)|
|Powering options||5V, USB|
|Weight||240 g (8 oz)|
|Dimensions||70 mm W x 89 mm L x 49 mm H|
|Communication interface||RS-232, USB|
|Operating||0° to 40° C (32° to 104° F)|
|Storage||-40° to 70° C (-40° to 158° F)|
|Operating Pressure and Altitude|
|Pressure||430 to 795 mmHg|
|Altitude||-380 m to 4572 m (-1,250 ft to 15,000 ft)|
|Storage Pressure and Altitude|
|Pressure||88 mmHg to 795 mmHG|
|Storage||-380 m to 15,240 m (-1,250 ft to 50,000 ft)|
|Operating and Storage||10% to 95% (non-condensing)|
|Standards and Regulations designed to meet:|
|ROHS Directive 2011/65/EU|
|BS EN ISO 21647 (including transport)|
|CSA C22.2 No. 601.1-M90|
|EN 60601-1-2 Emission Class B|
|EN 1789 (applicable parts for medical devices)|
|ISO 7137/DO-60 Emission Category M|
|MIL-STD 810 Vibration Method 514.5 Helicopter Category 14|
|IEC 60529 IPX2|
|Meets all applicable standards and regulations required to support a host monitor 510(k) submission.|