Optimized stability layer for more predictable deployment.
Redesigned tip and increased capsule flexibility for more dilator-like access and to better accommodate the most challenging anatomies.
Single-spine shaft for greater flexibility and improved deliverability.
Enhanced visualization. Radiopaque markers provide a reference for deployment depth and commissure location.
Performance as compared to Evolut™ PRO+ system in bench testing. Bench testing may not be indicative of clinical performance.
The Evolut FX stability layer increases catheter support during deployment and provides
Evolut FX conforms to the native anatomy more easily for a
The Evolut FX system has a(smoother) into simulated vasculature with a more flexible capsule.*
Evolut FX nosecone
Evolut PRO+ nosecone
The Evolut FX system has ain challenging anatomies.*
The Evolut FX systemfor greater control.*
*Performance as compared to Evolut PRO+ system in bench testing. Bench testing may not be indicative of clinical performance.
Tang, et al., found the commissural alignment technique allows for more accurate placement and orienting the hat marker at the outer curve during deployment reduces incidence of severe coronary overlap.2
Using the valve features of the Evolut™ FX system, and following Evolut procedural technique, commissure alignment can be achieved.
Radiopaque markers are a visual reference for TAV commissure location and alignment during deployment.
Existing cusp overlap implant views provide standard anatomical reference point to identify native commissure.
Tarantini G, Fovino LN, Scotti A, et al. Coronary Access After Transcatheter Aortic Valve Replacement With Commissural Alignment: The ALIGN-ACCESS Study. Circ Cardiovasc Interv. February 2022;15(2):e011045.
Tang GHL, Sengupta A, Alexis SL, et al. Conventional versus modified delivery system technique in commissural alignment from the Evolut low-risk CT substudy. Catheter Cardiovasc Interv. February 2022;99(3):924-931.