The Azure™ DR MRI and Azure SR MRI SureScan™ systems are indicated for rate-adaptive pacing in patients who may benefit from increased pacing rates concurrent with increases in activity. Accepted patient conditions warranting chronic cardiac pacing include symptomatic paroxysmal or permanent second- or third-degree AV block, symptomatic bilateral bundle-branch block, symptomatic paroxysmal or transient sinus node dysfunctions with or without associated AV conduction disorders, or bradycardia-tachycardia syndrome to prevent symptomatic bradycardia or some forms of symptomatic tachyarrhythmias. The Azure DR MRI devices are also indicated for dual chamber and atrial tracking modes in patients who may benefit from maintenance of AV synchrony. Dual chamber modes are specifically indicated for treatment of conduction disorders that require restoration of both rate and AV synchrony, which include various degrees of AV block to maintain the atrial contribution to cardiac output, VVI intolerance (e.g., pacemaker syndrome) in the presence of persistent sinus rhythm, or vasovagal syndromes or hypersensitive carotid sinus syndromes. Antitachycardia pacing (ATP) is indicated for termination of atrial tachyarrhythmias in bradycardia patients with one or more of the above pacing indications.
MRI Conditions for Use: Medtronic SureScan pacing systems are MR conditional, and as such are designed to allow patients to undergo MRI under the specified conditions for use. Pacemaker SureScan system patients may be scanned using a horizontal field, cylindrical bore, clinical 1.5T or 3T MRI system for hydrogen proton imaging. When programmed to On, the MRI SureScan feature allows the patient to be safely scanned while the device continues to provide appropriate pacing. A complete SureScan pacing system, which is a SureScan device with appropriate SureScan lead(s), is required for use in the MR environment. To verify that components are part of a SureScan system, visit mrisurescan.com. Any other combination may result in a hazard to the patient during an MRI scan.
The Azure DR MRI and Azure SR MRI SureScan systems are contraindicated for concomitant implantation with another bradycardia device or with an implantable cardioverter defibrillator. Rate-responsive modes may be contraindicated in those patients who cannot tolerate pacing rates above the programmed Lower Rate. Dual chamber sequential pacing is contraindicated in patients with chronic or persistent supraventricular tachycardias, including atrial fibrillation or flutter. Asynchronous pacing is contraindicated in the presence (or likelihood) of competition between paced and intrinsic rhythms. Single chamber atrial pacing is contraindicated in patients with an AV conduction disturbance. ATP therapy is contraindicated in patients with an accessory antegrade pathway.
Changes in patient’s disease and/or medications may alter the efficacy of the device’s programmed parameters. Patients should avoid sources of magnetic and electromagnetic radiation to avoid possible underdetection, inappropriate sensing and/or therapy delivery, tissue damage, induction of an arrhythmia, device electrical reset, or device damage. Do not place transthoracic defibrillation paddles directly over the device. Use of the device should not change the application of established anticoagulation protocols. Patients and their implanted systems must be screened to meet the following requirements for MRI: no lead extenders, lead adaptors, or abandoned leads present; no broken leads or leads with intermittent electrical contact as confirmed by lead impedance history; the device must be operating within the projected service life, and the system must be implanted in the left or right pectoral region.
Potential complications include, but are not limited to, rejection phenomena, erosion through the skin, muscle or nerve stimulation, oversensing, failure to detect and/or terminate arrhythmia episodes, acceleration of tachycardia, and surgical complications such as hematoma, infection, inflammation, and thrombosis. Potential lead complications include, but are not limited to, valve damage, fibrillation, thrombosis, thrombotic and air embolism, cardiac perforation, heart wall rupture, cardiac tamponade, pericardial rub, infection, myocardial irritability, and pneumothorax. Other potential complications related to the lead may include lead dislodgement, lead conductor fracture, insulation failure, threshold elevation, or exit block. Potential MRI complications include, but are not limited to, lead electrode heating and tissue damage resulting in loss of sensing or capture or both, or induced currents on leads resulting in continuous capture, VT/VF, and/or hemodynamic collapse.
See the device manuals before performing an MRI scan for detailed information regarding the implant procedure, indications, MRI conditions of use, contraindications, warnings, precautions, and potential complications. For further information, call Medtronic at 1-800-328-2518 and/or consult the Medtronic website at medtronic.com.
Caution: Federal law (USA) restricts these devices to sale by or on the order of a physician.